The emissions of wood-burning stoves and pellet stoves has been much in the news in recent years. It is said that wood-burning stoves emit a lot of particulate matter and pellet stoves actually emit a lot of ultra-fine particles. However, very few publications can be found where ultrafine particles have been measured. We had the idea of measuring ultra-fine particles ourselves which has now become possible (using an ultra-fine particle monitor). In this blog we describe our results and experiences gained during a day of measuring on an unique location.
2. Ultra-fine particles
Ultrafine particles (UFP) are the smallest particles in the air with a diameter from a few nanometers to 0.1 micrometer (<0.1µm). These particles are so small that after inhalation they end up directly in your bloodstream and can cause damage throughout your body.
3.1 Ultra-fine particle monitor
We measured the concentration of UFP with the CPC 3007 from TSI. CPC stands for condensation particle counter. The monitor uses iso-propyl-alcohol to solvate the UFP particles so that they become large enough to be measured by the laser. The monitor measures the total number of UFP particles with a diameter of 10 nanometers to 1 micrometer. The unit of UFP concentration is particles/cm3.
3.2 Measuring location
We have been able to find a measuring location where we can measure wood-burning stoves and pellet stoves. The measurements can be taken in the living room, but also at the outlet of the chimney. It is possible to measure different types of stoves at this measuring location. In this instance we opted for a computer-controlled pellet stove from 2019 and a traditional wood-burning stove from 2015 (a non ECO-design model).
4. Results & Discussion
4.1 Pellet stove
In the first instance, we measured UFP at a computer controlled pellet stove in the living room. After starting the pellet stove, the concentration of UFP in the living room was monitored for half an hour. The measurement was carried out at a distance of 3 meters in front of the pellet stove in a living room of 25 m2. After starting the pellet stove, smoke was visible in the combustion chamber. This smoke partially entered the living room and caused a temporary increase in the UFP concentration of approximately 11000 to 19000 particles/cm3, following that the UFP concentration remained between 9000 and 13000 particles/cm3.
Following the indoor measurement, the UFP concentration was measured at the outlet of the chimney at a distance of 25 cm. The outside UFP concentration of the clean air was approximately 7500 particles/cm3. The measurement at the chimney showed that peak values in the UFP concentration went up to almost 600,000 particles/cm3! These are extremely high values. According to the Dutch National Institute for Health and the Environment, a concentration of 100,000 particles/cm3 should be considered as an extreme value and is achieved, for example, at a very busy motorway in a large city.
4.2 Wood stove
After the pellet stove it was it was now the turn of the wood stove. The wood stove is an 8 kilowatt model from 2015. Before we started the measurement on the wood stove, we ventilated the living room by opening two windows and waiting for the UFP concentration to stabilize to the background level.
The wood stove was started according to the Swiss heating method. After two logs have been added to the fire, the UFP concentration in the living room started to increase considerably. The fire became so hot that the air supply to the combustion chamber was needed to be reduced. From that moment on, the UFP concentration increased further to around 150,000 particles/cm3. After 15 minutes two more logs were added to the fire. The catalytic afterburner ABCAT was then switched on and from that moment no more logs were added to the fire. We stopped measurement in the living room for half an hour in order to carry out measurements at the chimney outlet of the wood stove. After the measurement at the chimney was finished, it was resumed in the living room.
During the burning out of the wood stove, the UFP concentration increased to a maximum of 325,000 particles/cm3. This is again an extremely high concentration. It became so hot in the living room that we decided to open two windows fully to ventilate. During ventilation, the UFP concentration slowly dropped back to the background level. The wood burning stove was not yet burnt out, a glowing ash residue was still visible.
Just like with the pellet stove, we also measured UFP at the chimney of the wood stove. We also tested the effect of different types of catalytic filters. This was a palladium catalyst from ABCAT, Poujoulat and an unknown Chinese brand. The measurement was performed at a distance of 25 cm from the outlet of the chimney.
The UFP concentration without catalyst was at a maximum of 505,000 particles/cm3. With the ABCAT switched on, the concentration increased to a maximum of 525,000 particles/cm3. With the Poujoulat catalyst, the concentration increased to a maximum of 560,000 particles/cm3. With the Chinese filter switched on, the concentration increased to a maximum of 490,000 particles/cm3. The measured values at the chimney of the wood stove are comparable with the values measured at the chimney of the pellet stove. All measured values are extremely high. It appears that the catalytic filter further increases the concentration of UFP. The Chinese filter caused more resistance in the flue gas channel, so that the UFP concentration was less high compared to the other filters.
We were extremely surprised by the measurement results from this first experiment. We expected that we would measure high UFP concentrations but we certainly did not expect that the levels recorded in the living room would be so high, we were shocked. The plan is to repeat this experiment with other types of pellet stoves and wood stoves. We hope to be able to measure on a modern ECO design wood stove next time. Our story continues!